Choose a free accessory with a multifunction tester.Shop Now
This simple-to-use, low cost vibration meter is ideal for recording, analysing and finding problems with the vibration of machinery and equipment.
Supplied as standard with a BNC cable accelerometer with built-in magnet, the TPI 9071 can be used to easily find a clear picture of problems with unbalance, misalignment and looseness. For more complex issues, an 800-line spectrum with zoom and cursor helps to identify complex issues.
The TPI 9071 is suitable for use in a wide variety of areas, including commercial/industrial HVAC, food processing, manufacturing, mining, packaging, petrochemical, pulp/paper production, power plants and utilities.
During testing, the TPI 9071 will give you an overall snapshot into the machine under test's machine/bearing conditions. Vibration values are displayed on the screen with colour-coded alarm levels for ISO values and bearing damage (BDU), with it possible to switch to unbalance/misalignment and looseness mode for a clear picture of problems.
The TPI 9071 is also a datalogging vibration meter and can store up to 200 readings in internal memory for easy recall.
|Size||200 x 60 x 26mm|
|Environmental||Water: IP67 waterproof
Operating: -20 to 70°C
Storage: -20 to 70°C
|Power Supply||2 x AA batteries|
|Battery Life||Approx 50 hours of normal use|
|Frequency Ranges||2/10Hz - 1kHz ISO
2Hz - 10kHz Total g
1kHz - 10kHz Bearing Analysis
|Frequency Resolution||100/800 lines|
|Displayed Amplitude Units||Acceleration in g
Velocity in inches per second or mm per second
Bearing damage in BDU
|Displayed Frequency Units||Hz, CPM or RMP|
|Input Range||±50g (with supplied accelerometer)|
|Dynamic Range||72dB (0.05g resolution)|
|Measurement Storage||200 readings|
|Accelerometer Connection||BNC connector cable with removeable magnet (1mV/g - 10000mV/g)|
We'd really appreciate it if you'd review this product
Why not leave a review of this product? Reviews help us improve our service for customers looking for a product to make a decision.No reviews have been found
Do you have a question or found it cheaper? Contact us