Choose a free accessory with a multifunction tester.Shop Now
High performance, high speed and affordable LCR meter suitable for use on production lines and as a benchtop-based instrument in laboratories.
Ideal for testing components, quality control and even for being integrated into automated testing procedures (needs component handler and system controller to work), the AT810 makes it easy to fully automate your component testing, sorting and quality control data processing.
Featuring multiple selectable testing functions (L-D, C-D, R-Q and Z-Q), adjustable testing signal frequency of 100Hz, 120Hz, 1kHz and 10kHz, adjustable signal level of 0.1V, 0.3V and 1V with high basic accuracy of 0.1%, the AT810 makes getting the measurements you need as simple as possible.
Please note: accuracy is defined as meeting all of the following conditions.
|L||0.01μH to 9999H|
|C||0.01pF to 9999mF|
|R, Z||0.0001Ω to 99.99MΩ|
|D||0.0001 to 9.999|
|Q||0.0001 to 999.9|
|θd||-179.99° to 179.99°|
|θr||-3.1416 to 3.1416|
|Measurement Speed||25 times/sec
|Output Impedance||30, 50 and 100Ω|
|Ranging||Auto, Hold range. Total 6 ranges|
|Equivalent Circuit||Series and parallel|
|Trigger Mode||Internal, manual, external and bus trigger|
|Built-in Interface||Handler interface and RS232 interface|
|Temperature and Humidity Range||15 to 35°C, 80% RH or less|
|Storage Temperature and Humidity Range||0 to 50°C, 10 to 90% RH
|Power Supply||AC 198 to 252V, 48.5 to 62.5Hz|
|Maximum Rated Power||30VA|
We'd really appreciate it if you'd review this product
Why not leave a review of this product? Reviews help us improve our service for customers looking for a product to make a decision.
Review by David (Posted on 02/09/2018)
Very usefull instrument, this is a 10khz model but it seem more precis than my handled one, i use from 3 years.
One of the best feature it has is the series or equivalent operation mode and last but not the list the voltage testing output regulation with 3 selectable values, in the end i'm very satisfied with this instrument.
Review by Mike (Posted on 09/01/2016)
Very pleased with the instrument itself, but sadly it is let down by the very poor quality of the user instruction manual supplied with it. The instructions are not clearly defined and there are items not fully covered in it.
Do you have a question or found it cheaper? Contact us