Choose a free accessory with a multifunction tester.Shop Now
Use the RapidTrace 1200 to quickly find user opens, shorts, splices, bridge taps, split/re-splits, wet splices, frayed cables, water ingression, taps and splitters on network cabling within seconds.
Suitable for use on a wide variety of cables, including telecom, cat (3-6), power cabling, coaxial and mains power cabling, the RapidTrace can measure across cables from a length of 0.5m to 6km with selectable impedance of 25, 50, 75 and 100Ω.
Up to 50 individual traces can be stored on the device's built in memory for comparison over a time period. Historical data (stored on PC) can also be uploaded to the tester to compare against current results.
This tester also includes a built-in tone generator, has a USB interface and is ideal for use by telecommunication companies, data/network cabling installers, CCTV/satellite TV installers and electricians working on metallic cable installations.
|Ranges||7m, 15m, 30m, 60m, 120m, 250m, 500m, 1km, 2km, 3km, 6km|
|Range Selection||Manual range control|
|Accuracy||1% of selected range|
|Resolution||Approx 1% of range|
|Sensitivity||Min 3 pixel return at 4km on 0.6mm ø, PE, TP|
|Velocity Factor||Adjustable from 1 to 99% or feet, meters/ micro second|
|Output Pulse||5 volts peak to peak into open circuit|
|Output Impedance||Selectable 25, 50, 75 and 100Ω|
|Output Pulse Width||3ns to 3ms, automatic with range|
|Scan Rate||2 scans / second or scan held|
|Memory Capacity||50 Traces|
|Tone Generator||810 to 1000Hz|
|Battery Life||60 hours continuous scanning|
|Power Supply||6 volts 4 x 1.5AA alkaline cells|
|Power Down||Selectable 1, 2, 3 minutes or disabled|
|Backlit Display||128 x 64 pixels|
|Voltage Protection||250V AC|
|Operating Temperature||-10 to 50°C|
|Storage Temperature||-20 to 70°C|
|Dimensions||165 x 90 x 37mm (6.5 x 3.5 x 1.5")|
We'd really appreciate it if you'd review this product
Why not leave a review of this product? Reviews help us improve our service for customers looking for a product to make a decision.No reviews have been found
Do you have a question or found it cheaper? Contact us